The new Satelec X-Mind™ DC generator is fitted with a Toshiba DG-073B-DC double anode tungsten tube (intensity of 4mA and 8mA).
It generates a constant voltage of 60kV or 70kV.
It operates at high frequencies and constant potential which enables very high quality radiation to be emitted, whatever the conditions of use.
The choice of voltage (60 or 70kV) and intensity (4 or 8mA) enables optimal diagnosis of intraoral X-rays with a 35% reduction in exposure time, compared with monophased units.
The timer can control up to two DC generators.
The extremely short exposure times (minimum 0.02 sec.) are suitable for digital radiology with CCD, CMOS or phosphor plate sensors (radiation is 50% less that that used with “E” type films).
The SOPIX series offers wide range of solutions adapted for all budgets to satisfy all requirements of any dental practice:
SOPIX digital sensor offers a high-quality image at a very affordable price.
It is the most economic solution of the SOPIX Series
The design of the SOPIX series sensors is so smart.
The corners are rounded to improve patient comfort.
White side stripes offer high visibility of the sensor in the darkness of the mouth.
They assist the dental professional in correctly positioning the X-ray tube perpendicular to the sensor.
Moreover, the working environment now becomes more ergonomic and well-organized through the integration of SOPIX² inside sensor into X- Mind unity intraoral X-ray system.
Placed on his holder, the sensor is safe to prevent from falling to the floor and it is always easy to reach for optimal working comfort.
Through SOPRO’s extensive experience in digital sensors, the SOPIX series improve your everyday life by simplifying use and bringing you exceptional image quality.
ACE® technology, patented by SOPRO, is integrated in all SOPIX series sensors. It analyzes in real time the amount of X-rays accumulated by the sensor and It freezes the image acquisition as soon as it received the radiation required to provide a good-quality image. Thus, It protects each image from overexposure.
Specifications of x-ray machine
Electric classification : Electromedical equipment, Class 1 type B
Classification according to 93/42/EEC Directive : Class IIb
Supply voltage : 230V - monophase 50/60Hz
Power absorption at 230V : 1.4kVA
Resistance of the line : 0.5 ohms
X-ray tube : New Toshiba DG 073B DC tube
High voltage : 60-70kV
Anode current : 4-8mA
Focal spot : 0.7mm
Total filtration : Equivalent to 2mm Al at 70kV
Tube inherent filtration : Equivalent to 0.8mm Al at 70kV
Leakage radiation : Less than 0.25mGy/hour maximum at a distance of 1m from focal spot
Long cone : Focus to cylinder tip distance = 31cm (12”)
Timer : Exposure time can be set from 0.02 to 3.2 seconds
X-ray emission control : “Dead man” switch with 3m spiral cable
Total weight : 25kg (weight of the head : 5.5kg)
Specification of RVG sensor
Technology : CMOS + scintillator + optic fibre
Pixel size : 20 μm x 20 μm
Theoritical resolution : 25 lp/mm
Two sensor sizes : Size 1 : 25 x 39 mm (1.50 MPixels)